Read or Download Agilent 16533A, 16534A Digitizing Oscilloscope (help volume) PDF
Best nonfiction_6 books
Evaluating Religions covers such very important issues as contemporary theoretical ways to comparability, case stories of evaluating religions within the school room, and the influence of postcolonialism and postmodernism at the modernist assumptions of comparitivism within the educational research of faith.
- Seventh-Day Adventism Renounced
- A Review of Criticality Accidents [LA-13638]
- Cosmological Constant [lg artticle]
- [Article] Anomalously enhanced photoemission from the Dirac point and other peculiarities in the self-energy of the surface-state quasiparticles in Bi2Se3
- Karyotypes of Parasitic Hymenoptera
Extra info for Agilent 16533A, 16534A Digitizing Oscilloscope (help volume)
It shows the current acquisition only. Accumulate draws subsequent waveforms in the same area, without erasing previous waveforms. The accumulated waveforms are erased if any settings are changed, however. Average averages the current acquisition with the specified number of prior acquisitions. All acquisitions are equally weighted. The averaged waveforms are replaced with the current acquisition if any settings are changed. Acquisition Memory to Display For lower time resolutions, these settings allow you to optimize the oscilloscope for greater detail or longer duration.
When Pattern There are five ways you can use to further qualify the pattern that you want to view. They are: NOTE: Entered When the scope finds the pattern, it triggers on the edge of the pulse that makes the pattern valid. Exited The scope arms the trigger circuitry when it has found the pattern and triggers on the edge of the pulse that ends the pattern. Present > The scope triggers when the pattern is found and is present for greater than the time value that you specify. Present < The scope triggers when the pattern is found and is present for less than the time value that you specify.
3. Connect both to the same display tool. 4. In the Correlation Error dialog that appears, select Group Run for the logic analyzer and the logic analyzer description for the oscilloscope. 5. A Trigger Advisory dialog box may appear. Select Trigger Immediate. 6. Select the Group Run or Run All button to start the acquisition. 7. To view the waveforms together, open the display tool. • For a Waveform display, select one of the labels and choose Insert before... or Insert after.... In the Label Dialog, select the label you want to insert, then select the Apply button.
Agilent 16533A, 16534A Digitizing Oscilloscope (help volume)